EIT PALM Probe UV Radiometer - Light Intensity Meter
The PALM Probe radiometer from EIT addresses UV Measurement and Process Control in areas where it has been difficult to pass a radiometer through the process. The use of UV inks and coatings has grown in label and web markets and the PALM Probe offers safe, reliable measurement for these and other applications.
The word PALM is an acronym for Production Ambient Light Measurement. The PALM Probe has an extremely wide dynamic range allowing it to read from very low levels of UV light as in fluorescent bulbs to very high levels of UV found in powerful UV curing systems. The physical conditions inside a UV curing chamber are harsh. An instrument must withstand wide temperature variations. A tremendous amount of electrical energy is used to excite the UV bulb. The PALM Probe is designed to withstand the harsh physical conditions of the UV curing chamber as well as protect the operator and instrument from damage or electrical shock. Potentially lethal voltages exist in UV curing systems. Although the Light Guide has been designed to reduce the risk of shock, the user should avoid contact with the high voltage areas in the UV housing. The user holds the Probe Body of the instrument and inserts the Light Guide under the UV source.Operation:
Ultraviolet, visible and infrared light radiation of all wavelengths impinge on the Input Aperture at the tip of the PALM Probe Light Guide. The light is directed down the Light Guide to the base of the Probe body where the UV filter passes the UV light of interest to the photodetector. The photodetector converts the light energy to a current that is proportional to its intensity. The signal is conditioned, digitized, processed and displayed on the LCD of the PALM Probe.
The Light Guide is constructed out of steel alloy and coated with a non-conductive ceramic coating to insulate and protect the user from accidental shock while taking measurements. The Light Guide is further isolated from the Probe Body by a non-conductive Delrin block. The Probe Body contains the optics, electronics, battery and display functions of the unit. The Probe Body can be held either with an 'over' or 'under' hand type of grip. The raised membrane switches are easy to operate with either hand.
The PALM Probe has a very wide dynamic range that automatically adjusts itself for the user. For example, the PALM Probe can measure light from approximately 100 microWatts/cm2 (µW/cm2) up to 10 Watts/cm2 (W/cm2). The decimal point in the LCD 'floats' over three places to cover the wide dynamic range on the unit.
The PALM Probe display allows the user to alternate between Watts-Joules-Seconds during data collection and after a reading or run has been made. The SELECT button allows the user to toggle between the values.
When collecting irradiance data with the PALM Probe, the display will change as the Probe changes position under the UV source. As the aperture opening moves into the focused peak irradiance area of the UV source, the reading will increase. Variations in irradiance may also be seen when the PALM Probe is moved along the length of a UV bulb. The LCD display will automatically scale and adjust as the PALM Probe takes readings. Pressing the RUN/STOP button will stop the collection of data and the highest irradiance value measured during the collection of data will be displayed. Most users will find that the Irradiance value (W/cm2) displayed is the most important value collected by the PALM Probe when it comes to process control for a particular application.
UV Energy Density or Dose incorporates time into the irradiance measurement. The PALM Probe displays the Energy Density or Dose value in Joules (J/cm2 ) on the LCD. The dose is an integration of the area under the irradiance curve. The dose reading increases as the reading continues. The speed at which the dose increases depends on the irradiance value. Switching to the built in timer in the PALM Probe will help monitor the 'time component' of the dose reading for more consistent results. Placement of the PALM consistent and repeatable irradiance values to calculate dose readings.EIT UV PALM Probe Users Manual (PDF)
The following diagram shows all the available LCD Displays on the PALM Probe. In operation, the display is easy to read and understand as the user will only see the segments on the display that pertain to current reading.EIT UV PALM Probe Brochure (PDF)
Features at a glance:
- Wide Dynamic Range (100,000:1) Measurements; Autoranging and zeroing.
- Measures and displays Watts, Joules, Seconds; displays peak value Watts/cm2.
- Electrically isolated and insulated Light Guide.
- Very high temperature resistant Light Guide.
- Single hand operation; two membrane switch control.
- UV bands: choice of EIT UVA, UVB, or UVV.
- Battery powered; two user replaceable AA batteries with low battery warning.
- Measure UV system performance particularly in applications where measurement space is limited or it is difficult to access.
- Establish, document and maintain UV process windows
- Meet ISO, quality and customer requirements for SPC/SQC.
- Verify readings from on-line sensors; coordinate relative percentage type readings from on-line displays to absolute units.
- Multiple uses from production curing (high irradiance) to stray hazard (low irradiance) applications.
- UV Range: 100 µW/cm2-10 W/cm2
- UV Spectral Response: UVA 320-390nm, UVB 280- 320nm, UVV 395-445nm
- Display: LCD with 4 digits and floating decimal point
- User Interface: Two raised membrane switches
- Operating Temperature Range: Probe Body 0-70 degrees Celsius
- Time-out period: 10 Minutes
- Batteries: Two AA Alkaline Batteries
- Battery Life: 25 Hours run time
- Mechanical Specifications:
- Overall Length: 25.75" (65.4 cm)
- Probe Body Length: 7.5" (19.0 cm)
- Light Guide Length: 18.25" (46.4 cm)
- Weight: 21 ounces (596 grams)
- Light Guide Temperature Resistance: 750 degrees Fahrenheit (400 degrees Celsius) on a continuous basis; much higher for measurement length exposures.
Questions & Answers
Have a Question?
Be the first to ask a question about this.